Bourahima Kone, Drissa Goita, Oumar Dolo, Daouda Traore, Dramane Sogoba, Amadou Somboro, Moumine Sanogo, Anou M Somboro, Nadie Coulibaly, Alou Sanogo, Zoumana Diarra, Madou Traore, Almoustapha I Maiga, Bocar Baya, Yeya Dit Sadio Sarro, Bassirou Diarra, Amadou Kone, Dramane Diallo, Djeneba Dabitao, Jane L. Holl, Michael Belson, Sounkalo Dao, Robert L. Murphy, Mahamadou Diakite, Souleymane Diallo, Seydou Doumbia and Mamoudou Maiga
Background: Routine monitoring of HIV-1 Viral Load (VL) is important in patients on Antiretroviral Therapy (ART) management. Access to HIV VL remains a challenge in resource-limited settings, especially in rural areas. Universal access to VL requires more simplified and less restrictive alternatives to current conventional VL methods. The objective of this study was to evaluate the performance of the new rapid (2-hour turnaround time) Xpert HIV-1VL technique compared to Roche TaqMan and Abbott RT m2000 for HIV-1 RNA quantification in HIV- infected patients.
Study design: We conducted a cross-sectional study in patients seen for routine VL monitoring between August and November 2018 in a HIV care site in Bamako. The performance of the Xpert HIV-1 VL assay was evaluated against the Roche TaqMan assay and Abbott m2000 RT assay. Performance, utility and reliability/reproducibility were verified using accuracy, sensitivity, specificity, positive and negative predictive values, Diagnostic Odds Ratio (DOR), Kappa coefficient, Pearson correlation coefficient, and Bland-Altman analysis.
Results: The Xpert assay compared well with the two current referral assays (Roche TaqMan and Abbott m2000 RT assays). Compared to Roche TaqMan assay the sensitivity was 93.10%, specificity (97.01%) and accuracy (95.20%), the correlation coefficient of Pearson (r) was 0.98 (p <0.01). Bland-Altman anal d a mean difference of 0.18 log10 cp/mL; (Standard Deviation) SD=0.33. Compared to the Abbott m2000 RT, the sensitivity, the specificity and the accuracy were respectively 93.44%; 92% and 92.65%. The Xpert HIV-1 VL assay showed a good correlation with a correlation coefficient of Pearson, r=0.99 (p <0.001). The overall mean difference in the HIV-1 VL values obtained by Xpert HIV-1 VL and Abbott m2000 RT assays was 0.08 log10 cp/mL; SD=0.30.
Conclusion: Xpert HIV-1 VL showed a good performance compared to Roche TaqMan and Abbott m2000 RT. With the rapid test results (less than 2 h) and ease of testing individual specimens, the Xpert HIV-1 VL assay could be an effective alternative for HIV VL monitoring in resource-limited settings.PDF
Share this article